|Home||» Products||» Fabric Testing Equipments||» Perspirometer Tester|
|IS: 971- 1983||Method of determination of colour fastness of textile material to perspiration|
|AATCC||Method of test for colour fastness to perspirometer.|
Shading speed of colored or printed fabrics against sweat is dictated by uncovering the fabric example to the activity of both soluble and acidic reagents while in contact with undyed contiguous fabrics is set between plastic plate under a settled load inside a broiler kept up at 37 C. The mechanical assembly utilized for this test is known as the perspirometer.
The stacking edge comprises of two steel plates between which the acrylic plates are put. The test example are sewed between suitable white fabric pieces and plunged in either antacid or acidic test arrangements. The composite example in set in the middle of any two acrylic plates. Various test examples can be stacked at the same time between diverse arrangements of acrylic plates by stacking them one over the other. Stacking is finished by keeping a standard dead weight on top of the upper plate's get together, which comprises of two parallel plates isolated by suitable repaying spring. The aide bar joined to top plate of this gathering is bolted while under burden and the heap evacuated. The spring adjusts for ensuing variety in burden because of variety in thickness of the test examples on drying.
The test is proceeded for the obliged time under the predetermined state of temperature. After this the examples are expelled from the casing and dried. The level of recoloring of the white bits of fabric and change in shading of the test examples are assessed utilizing Gray scale and looked at against standard rating card. The whole parts are made of mellow steel and are brilliant chrome covered to give them a consumption safe completion. These items are produced using stainless steel can likewise be supplied against specific request.